This User’s Manual describes how to use the E5EK compact, high-function digital Nevertheless, OMRON assumes no responsibility for errors or omissions. E5EK-AA Omron Automation and Safety Controllers 1/8 DIN VAC Digital field config datasheet, inventory, & pricing. This product was manufactured at OMRON Okayama. OMRON Okayama has obtained approvals from international certification bodies for its quality system and.
|Published (Last):||2 May 2005|
|PDF File Size:||9.32 Mb|
|ePub File Size:||18.47 Mb|
|Price:||Free* [*Free Regsitration Required]|
When a potentiometer is not connected to the controller,  is dis- played on the No. Be sure to turn on the power supply to the load either before or simultaneously with the start of Temperature Controller opera- tion. E5EK controller has been correctly calibrated.
For details of operation at an alarm, see page About The Communications Function 1. In this mode, you can The parameter default is [ 4 As the setting in this example is to be left as it is, press the key.
AT auto-tuningsetting the alarm values, set- ting the control period, and setting PID parameters. Al- ways heed the information provided with them.
When replacing the e5sk unit, use the ER. In this case, omrron is carried out with the process value regarded as the set point before the change was made. Specifications Assign these output functions to control output 1, control output 2, auxiliary output 1, and auxiliary output 2. Failure to heed precautions can result in inju- ry to people or damage to the product.
Page 76 SP ramp function is enabled. Dead band Hysteresis heat. However, be sure to execute motor calibra- tion after connecting the potentiometer. Of these, only the alarm assigned as the output can be used.
Alarm output functions are activated when the upper limit is exceeded. Dead band Hysteresis heat Page The setting range varies according to this output data. LBA, 7 to 9: F Meaning Calibration data is in error. G5-series with built-in ethercat communications linear motor type pages. In an open status, only control output 1 is ON. However, if the E5EK controller must be calibrated by the user, use the parameters provided for user to calibrate temperature input, w5ek input voltage, current and transfer output.
Page 73 If this mode is o,ron to during auto-tuning, auto-tuning is canceled.
Omron E5ek, Omron E5ek Suppliers and Manufacturers at
Manual Operation F Dead band The dead band is set with the set point as its center. Also, a lower limit value larger than the upper limit value may be set.
Control output cool2: Page setup mode can be used. Don’t show me this message again. F Action Must repair. However, when actually creating programs, do not leave spaces between frame items. Page 78 AT auto-tuningsetting the alarm values, set- ting the control period, and setting PID parameters.
Omron E5ek-aa2-500 Digital Controller Plug in Cover (cover Only)
Set- ting a positive value produces a dead band, while setting a negative value produces an overlap band. When the value on the No.
Function PV before passing through filter PV after passing through filter 0.